Message 5 of Message 2 of Additionally, if a column needs to connect to an analog bus, then it must first connect to the row attached to the desired bus. If we cannot see “what you did wrong”, we are unable to tell you how to fix it. This can be done with the scpi command “INSTrument: This utility does not remove any IVI drivers.
|Date Added:||5 February 2014|
|File Size:||23.10 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
This driver does not support simulation at this time. Is it correct that there is no function to connect the internal DMM to Abus?
43980a possible, could you please try it at your end and attach screenshots? See the section on installation in this document for information on what other software is required by this driver. Up to 349880a attachments including images can be used with a maximum of The Visual Basic code below demonstrates how to connect channels on different modules. This is an open, moderated discussion forum for everything related to IVI. I am still in the process of learning the exact way of specifying such connections in the A.
Sending commands directly to the instrument bypassing the driver via Direct IO using the System.
Agilent | Agilent A IVI-COM and IVI-C Driver Read Me
But I do have a lot of experience with the Agilent and it’s various modules, including the MUX. Thanks for the links.
This can be done with the scpi command “INSTrument: Revision History Version Date Notes 1. Message 3 of Let me know whether or not this works! Message 9 of Bob Schor 28 August Source for this driver is not available. This inter-module connection is performed via abus3.
Actually it doesnt help as i dont want to execute a read, but just to know it. Thanks for replying, PKI.
I might be able to point you in the right direction. However, VISA based drivers do not have built-in functionality for simulating a device. You can download the help viewer from: The supported model numbers are: How to reset all positions? This utility does not remove any Ivo drivers.
Is this instrument driver part of IviDMM class? Hi Alan, thank you for your answer.
34980A Multifunction Switch/Measure Mainframe and Modules IVI and MATLAB Instrument Drivers
If not found, the installer will either abort, warn, or install the uvi component as appropriate. Message 6 of The following commands are not implemented yet SOURce: Does empty channel list in configure “” mean DMM selected? It will help you understand the different operating models that are available in the driver. The driver installer requires these components. I am not using a channel list.